Both bit and bit Windows runs bit applications properly. Trigger Level too high, and will be changed to One frame is equivalent to one sample of each active channel. Failed to send the unlock key to some older DATAQ devices or failed to send out the scanlist to new devices. If you need to enable channels with different combination, you can use UltimaSerial. A trigger event will be fired when the trigger condition is met. GetData ensures waveform data continuity by passing a continuous stream of data to your program.
|Date Added:||14 October 2004|
|File Size:||49.61 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
All Valid Trigger Channel: PGain index changes P rogrammable Gain table. This method must be invoked after Start Applicable Devices: Sorry, but the pattern of the square wave cannot be controlled. A trigger event will be fired when the trigger condition is met. PreTriggerLength should be zero Applicable Devices: This channel does not support alarm setting Each channel can have di-194gs limit lines, one for high limit and one for low limit.
One frame is equivalent to one sample from every active channel. NET Class if you are starting a new project on products supported by the class. One frame is equivalent to one sample of each active channel.
If you enter a number that is higher than the number of possible channels for the dtaaq acquisition instrument you are using, ChannelCount will be reset to 1. There are multiple functions for this property.
For examples in other programming languages, please visit our classroom. Invalid device name The actual sample rate can then be read after you run Start method.
Data Logger Starter Kit
Original channel count is unmodified To catch the Maximum reading of a specified channel in the duration of the test from the moment Start method is invoked to the moment of this method is invokedadd to the valid channel count mentioned above. This function reports how many available new data points have filled the buffer since the last time data was retrieved using GetData.
Fired when the number of new data points are equal to or higher than di-14rs is specified in the property of EventLevel. Default value is GetData returns waveform data as a variant of two dimension bit integer array. MaskOutDigitalBits Boolean when set, the digital bits will be masked out from the data stream default: Channel count is changed to 1 Count is an integer and defines the number of scans you are requesting.
The event NewData fires whenever a set number, or more, of new data points have been gathered.
Channel 8 is always digital input, Channel 9 is always Rate input, and Channel 10 is always a bit Counter input. Specify the COMM port dattaq port you want to use or find out which one you are currently using. The included samples are mostly from VB6.
When Ultimaserial is first loaded, the channel allocation table is 1, 2, 3, NET is providing an advanced. If you try to access the digital channel which is not available on the device, the result is meaningless see below for further explanation.
Dataq Instruments Di-194rs 4 Channel Data Acquisition Module & Cable
The EventLevel property allows you to specify when you want the NewData event fired. It serves as the entry point for the superkeys of site license Example: To catch the Minimum reading of a specified channel in the duration of the test from the moment Start method is invoked to the moment of this method is invokedadd to the valid channel count mentioned above. Lets you write the number of channels you want to be enabled or read the number of channels that are currently enabled.